October 1, 2020
Radiofrequency (RF) electrosurgery if utilized properly, it can permit the specialist to play out a wide range of systems securely, successfully and with negligible undesired tissue injury. Utilized without appropriate consideration, instruction, and preparing, electrosurgery, as different instruments and vitality sources, can possibly cause unreasonable tissue injury, and expanded employable dismalness, now and again of a perilous sort. This section furnishes the peruser with a basic information on the logical standards of RF electrosurgery that will encourage comprehension of the frameworks and procedures.
While setting up Operating Room for electrosurgical gadgets, one should follow some precautionary measures.
State of being(Equipment Condition)
— Analyze the ESU and its adornments for surrenders—don’t utilize links or frill with harmed (split or taped) protection or connectors.
— Affirm that the ESU has been assessed for security and execution by a certified engineer.
Return Electrode Monitoring (REM) or Link Progression Alert
— Check the activity of the REM or the return electrode monitoring by endeavoring to work the unit with the dispersive terminal separated—the unit will get activated and an audible tone should sound.
Test Unit Before Using it on the patient.
—All gear ought to have normal support checks and tests to guarantee they are working appropriately. Never use hardware on a patient until appropriate security tests have been played out, any important changes have been made, and the piece has been re-tried.
Audible Activation Indicator
— Activate the unit using each footswitch or hand pencil, and can verify that the audible activation tone is loud enough to be heard over other noises inside the Operating Room.
— Check activity of some other cautions or defensive highlights.
Use Equipment and accessories as they were intended to be utilized.
—Just following to the guidelines and safeguards gave all gear will extraordinarily decrease the chance of injury. Use generators at the least setting suggested for the technique and never substitute adornments and use them in some other strategy than for what they were planned.
May 26, 2020
Infants / newborns basically have to handle any of these two phases one is hyperthermia it is characterized by an uncontrolled increase in body temperature that exceeds the body’s capacity to lose heat, in contrast to a fever. Causes of hyperthermia in infants include exposure to a hot environment to tackle this pediatrician uses Phototherapy unit.
The Second Phase is hypothermia which is the critical one Hypothermia (decrement of body temperature ) at birth is one of the most important risk factors for newborn morbidity and mortality. 99% of newborns that die globally are in such settings. It is important to keep infants warm and help them achieve thermoregulation in order to prevent and minimize morbidities and mortalities due to hypothermia. The device used for this is Radiant Warmer / Incubator.
Infant Radiant Warmer, is a body warming device to provide heat to the infant body. This equipment helps in managing the body temperature of the baby and limit the metabolism rate. Heat has a tendency to flow in the heat gradient direction that is from high temperature to low temperature. The heat loss in some newborn babies is fast therefore body warmers provide support to keep the body temperature constant.
Warming Device Consist of two modes – Manual & Servo
Radiant Warmers consists of a baby bed(where the baby is kept) and the artificial heating is provided by a heating mechanism mounted overhead. The heating mechanism can be of quartz, caloroid or ceramic which produces desired heat and a reflecting mechanism(reflector above heating element) helps to divert the heat at the baby bed. The skin temperature of the baby can be monitored by a temperature measuring sensor that is kept continuously attached to the body and one air sensor for additional safety of the baby as the sensor will cut the heater output if the air temperature crosses the desired temperature. The variation in the skin and the temperature can be seen on a small LCD panel which continuously shows the body temperature. Radiant warmers are equipped with alarms and APGAR timer to indicate the change in temperature and hence attract attention of medical professional attending the baby. The heat generated can be controlled manually(Manual mode) by a knob or digital switch as well as automatically(Servo mode) depending on the Radiant Heat Warmer setting.
The Radiant warmer can be used as servo or manual depending on the baby condition and operator. in servo mode a pre defined temperature is set 36.5* C the temperature will rise gradually and will cut once it reaches to the desired temperature and will try to manage the temperature. whereas in manual mode user have to define the heater power and temperature as well for how long the device has to be activated.
It is always recommended to have a proper training and read the manufacturers guidelines for person handling this equipments. It is necessary to regularly clean and disinfect the instrument.
May 26, 2020
What is Physiotherapy?
Physiotherapy is a significant piece of treatment for the vast majority with joint inflammation.
A physiotherapist are the part of healthcare professionals and helps to manage and improve with physiotherapy devices and gadgets when the technological advancements weren’t that exceptional, we used to go for conventional and traditional approaches. But today we have come up with right technology and devices.
A task of physiotherapist –
To provide or offer advice and reassurance and helps the patient to feel confident about managing condition.
To address any concerns or uncertainties and will set appropriate goals to keep you as active as possible.
MEDILAP Since the foundation of Medichem Electronics Pvt Ltd in year 1998, the Company is engaged in manufacturing of a wide range of Electro Medical devices and a whole range of equipments for pain management, to rehabilitate the patients for post Operative care.
MEDILAP is always keen in helping physiotherapist in redesigning their career as a Professional Therapist and start a physiotherapy clinic with affordable price and best quality.
Quality assures reliability and guarantee
(Initiating any exercise starts slowly and increases in small steps. This helps to strengthen muscles and joints and increase fitness. Improving general fitness and stamina will help to increase activity level without increasing pain.)
Electrosurgery is the application of a high-frequency (Rf) alternating polarity, electrical current to biological tissue as a means to cut, coagulate, dessicate, or fulgurate tissue particle.
ESU’s (Electrosurgical unit), by contrast, uses radio frequency (RF) alternating current to heat the tissue by RF induced oscillation at a frequency more than 300 KHz of molecules that results in increase of temperature and produce heating to a certain level which produces heat to the tissue, instantaneous cell death occurs.
In fact, ESUs are used in more than 80 percent of all surgical procedures, including veterinary.
The difference between Scalpel based surgery and Modern day ESU (Electrosurgery) is traditional surgery causes a heavy blood loss and electrosurgery prevents the patient from excessive bleeding which can cause a big impact. Reduced risk of hemorrhaging and a precise surgery which helps in quick recovery.
Medichem Electronics Pvt Ltd has been Manufacturing (Brand MEDILAP), designing and developing using advance micro controller based technology for the last twenty years.
RF electrosurgery is performed using a Surgical Diathermy) (also referred to as an electrosurgical unit or ESU) and a handpiece including one or two electrodes, a monopolar or bipolar instrument and accessories. All RF electrosurgery is bipolar so the difference between monopolar and bipolar instruments is that monopolar instruments comprise only one electrode while bipolar instruments include both electrodes in their design.